Optical windows for combustion research and control applications: Anti-fouling strategies
Authored by: Helmut Ranner, Maximilian Lackner, Heinrich Kofler, Franz Winter.
Affiliation: Institute of Chemical Engineering,
Vienna University of Technology,
Getreidemarkt 9/166, A-1060 Wien, Austria
To send an email click here.
As the authors of this paper say, the windows in a combustor play an important role in any optical experiments. This paper deals with the spectral and average intensity losses due to combustion deposits on optical windows. Anyone contemplating using any optical techniques in combustion devices should read this paper to minimize the losses due to window deposits.
Optical diagnostics for real-time combustion control, especially based on emission spectroscopy, have gained considerable importance, both with power plants and internal combustion engines.
A key element is the optical window. It must withstand the harsh and transient conditions in the combustion chamber and also show a low propensity to deterioration and fouling. Combustion chamber deposits (CCD) may rapidly weaken and eventually destroy the transparency of the window, hence inhibiting online optical diagnostics.
This work is about the experimental investigation of different window substrates and their mounting possibilities in combustion chambers. A specially prepared internal combustion engine running on liquid fuels was used. The window was mounted in the cylinder head next to the spark plug. The infrared-transparent media quartz, sapphire, ZnSe and ALON™, which are commonly used window materials, were compared experimentally. The influence of the window temperature and the effect of various fuels (gasoline, MTBE and isooctane) and of the lubricant were studied. The initial phase of deposit formation was investigated. Gravimetric analysis, transmission measurements and elementary analysis of the deposits are discussed.
It was found that the window temperature is the main parameter which influenced the deposition rate of combustion residues. Carbonaceous deposits were found to cease above 350°C outer window temperature. CaSO4, stemming from calcium sulfonate in the lubricating oil, was found to be the most important inorganic constituent in the deposits formed on the windows. Under transient operational conditions, the kind of fuel mixture as well as the substrate can determine the transmissivity of the windows.
Strategies of how to prevent fouling of optical windows in combustion chambers are discussed..
Combustion chamber deposits (CCD), window fouling, internal combustion engine, optical diagnostics
The full paper may be downloaded from the server, in the “New Papers” section (http://www.journal.ifrf.net/articles.html), by clicking on the Acrobat PDF icon alongside the title.
Publication in the Journal
The Editor-in-Chief would like to remind all potential authors that publication in the Journal is open to all. If you have interesting results to publish in the field of, or related to, industrial combustion, we invite you to prepare a paper according to the guidelines given in the Author’s Guide on the website (http://www.journal.ifrf.net/).
Papers may be regular “articles” (typically up to 20 pages) or Communications (typically up to 4/5 pages). Review papers can of course be longer. Remember that figures and graphics in general can be in full color. This advantage should be encouraged.
All manuscripts and associated files, proposed for publication should be sent by the Corresponding Author in a compressed/zip file, as an email attachment to firstname.lastname@example.org. This file should include a statement that the proposal’s content is unpublished material that has not being submitted for publication elsewhere. When an article by the author(s) is cited in the proposed article as “in press”, a copy of this article should accompany the proposed article and should be included in the compressed file.
The Editor-in-Chief looks forward to receiving your proposals.